Ask any non-NVH related questions about using PicoDiagnostics software here.
Are you sure your lead is ok and conected to good ground? Seems like ground issue.
- Posts: 183
- Joined: Fri Jan 03, 2014 3:43 pm
- Location: Estonia
judging by the time scale that trace is mains interference for sure 50-60 Hz . its definitely not an O2 sensor signal
- Posts: 5
- Joined: Wed Apr 28, 2010 6:33 am
Return to PicoDiagnostics software
Who is online
Users browsing this forum: No registered users and 0 guests