Hello Pat, I have just created a custom probe using the 60 amp current clamp in 60 amp mode. This will allow you to measure +- 60 amps on one scale. When editing a pre-set “automotive” probe click on the 60a Current Clamp (60A mode), then click “Duplicate”. This will duplicate the selected current clamp and place the clamp into the probe “Library”. Here you can then “edit” the settings for this probe with no fear of editing the settings of the original clamp. Scroll through the custom probe wizard until you arrive at the “Scaling Method” box. Click on the “Edit look up table” and erase the -50 value and type in -600, erase the -5 value and type in -60. Click OK and follow all the prompts to the end to click “finish”
Select this new probe from the probe library section under the channel options menu and select the relevant scale required. +-60 amp will be one of the options.
I hope this helps but please feedback if you need any further assistance.
- 60A Current Clamp [60A Mode] (2).psprobe
- (1.32 KiB) Downloaded 122 times